Officially licensed by the General Department of Medical Equipment
X-ray socket (X-RAY 32A) operating room is used to connect the radiology device in the operating room and is one of the requirements of operating room design.
For the mobile x-ray imaging machine, there should be one phase and neutral electrical outlet with ground connection, 250 volts, 63 or 100 amps depending on the need, which is fed from a separate isolated electrical panel in the operating, delivery, and fracture rooms. , be predicted.
Fixing Hole Spacing
Erath pin position
|IP44||83×90 mm||70×76 mm||Screw||250V||6h/200||CE||32A|
The thickness of the built-in box
|22.5cm||16cm||72cm||15.5cm * 12cm||18 months replacement warranty|
X-ray outlet facilities
|Has a washable stainless steel plate||Washable and steel container||Standard socket with European quality||Equipped with a protective fuse with a maximum passing current of 52 amps|
Lock the door
|Built-in steel box|